As the scale of the company continues to expand, environmental laboratories sometimes need to relocate. In the relocation process of the constant temperature and humidity test chamber, only the following three steps can be taken to complete the transfer work.
The constant temperature and humidity test box should avoid excessive tilt during the transfer process, and the maximum tilt angle cannot exceed 30 degrees. The refrigeration principle is the same as that of the refrigerator. The refrigeration oil and the ambient refrigerant in the compressor will flash out when excessive tilt occurs, resulting in the compressor Damage, this is a taboo for the industry, we must be careful.
The constant temperature and humidity test chamber must be packed with flexible materials such as foam or sponge during transportation to avoid appearance damage. The casters need to be fixed. After reaching the new laboratory, do not use it immediately because the constant temperature and humidity test chamber is in the process of transportation. The refrigerant and refrigerating oil are subject to vibration and require a certain amount of time to recover. It can be placed in a horizontal position for 12 hours.
After the constant temperature and humidity test box arrives in the new laboratory, the power is turned on and the switch is turned on. Sometimes, the device does not light up. This is because the test box is equipped with a phase sequence protection switch and only two of the three fire wires are required. Root can change the wiring position.
Through the above three steps, you can easily realize the transfer of the constant temperature and humidity box. The above was contributed by the Yashilin Technology Department. Please note the source.